Refurbished Eschmann TD850 Diathermy Unit
Pre-owned, working and in good condition. Three independent generators Unique multi-user facilities Specialist modes Unrivalled cutting performance Enhanced safety features.
The knowledge and experience gained in developing the renowned TD411 series have been applied to the TD850, creating one of the most technologically advanced electrosurgery units available today, offering three truly independent outputs.
Can be supplied with these accessories:
– >Monopolar blue and yellow footswtiches.
– >Bipolar footswitch
– >Bipolar cable.
– >Bipolar Forceps
– >Reusable rubber plate and cable.
– >Finger switches.
Refurbished TD850 diathermy powers up and passes its initial self-test.
The LED displays and indicators are fully functional.
Output was verified using a diathermy tester for all the channels and all modes including Monopolar cut, blend, specialist, pinpoint, spray and bipolar high & low coag.
Bipolar Cut could not be tested as no footswitch present.
All the values were within acceptabe limits.
Bipolar Coag footswitch is functional.
Monopolar Cut & Coag footswitches are working fine.
Resistance check was not carried out on patient plate monitoring circuit but it appears to be in good working order.